As of March 25, 2013, LifeScan has begun a voluntary recall and replacement of all of its OneTouch® Verio®IQ blood glucose meters in the United States. The meters are being recalled and replaced because of a technical problem that fails to deliver important information about extremely high blood glucose levels.
LifeScan has announced the voluntary recall of its OneTouch Verio IQ Meter in the United States and Canada. The company reports that under very specific circumstances, the meter turns itself off when users attempt to access the "Results Log" to view stored test results. When the meter is turned back on, it enters the "set up" mode and requires users to confirm the date and time settings before being able to test again.
According to LifeScan, this issue has no effect on the accuracy of test results or functions such as on-screen pattern notifications, averages, result tagging, or downloading, and data are not lost. If the user does not confirm all settings, however, testing is not possible. As a result, treatment may be delayed or a treatment decision made without a blood glucose reading.
Diabetes Health is the essential resource for people living with diabetes- both newly diagnosed and experienced as well as the professionals who care for them. We provide balanced expert news and information on living healthfully with diabetes. Each issue includes cutting-edge editorial coverage of new products, research, treatment options, and meaningful lifestyle issues.
The information on this site is not intended or implied to be a substitute for professional medical advice, diagnosis or treatment. All content, including text, graphics, images, and information, contained on or available through this website is for general information purposes only. Opinions expressed here are the opinions of writers, contributors, and commentators, and are not necessarily those of Diabetes Health.
Never disregard professional medical advice or delay seeking medical treatment because of something you have read on or accessed through this website.