You can view the current or previous issues of Diabetes Health online, in their entirety, anytime you want.
Click Here To View
Latest Insulin Pumps Articles
Popular Insulin Pumps Articles
Highly Recommended Insulin Pumps Articles
Send a link to this page to your friends and colleagues.
Troubleshooting the pump system involves looking at your pump screen to make sure the appropriate basal is set, the appropriate bolus was given at the last meal and the time is correct. If all of this is working appropriately, make sure your syringe has insulin in it, and there is no leakage. Make sure the set is attached appropriately to your body, and the needle or infusion site area looks normal.
If there is any leaking, you should change out your syringe and tubing. If there is leaking or significant air in the tubing, you should also change out your tubing and syringe.
If your infusion site is red or irritated in any way, remove the needle set and place a new one in another area under sterile technique, ideally prepping your skin first with Hibiclens prior to needle insertion.
If your infusion set is crimped upon removal, you need to read the instructions on how to appropriately insert the set.
If you have a hematoma or bleeding under the skin at your infusion site, this is probably not significant unless your glucose is high or your skin is red and warm around the infusion site. Redness and warmth would suggest you have an infection.
If you find there is some other mechanical problem with the pump, you should call the 800 number on the back of the pump and ask for further help.
Diabetes Health is the essential resource for people living with diabetes- both newly diagnosed and experienced as well as the professionals who care for them. We provide balanced expert news and information on living healthfully with diabetes. Each issue includes cutting-edge editorial coverage of new products, research, treatment options, and meaningful lifestyle issues.